They are embedded inside the instrument driver selections. Message 1 of Then I can get the curve U t and I t. You surely know the GPIB-adress of your instrument and the number of your bus while it can sometimes be difficult to get a correct reference. Thank for your help. Do you want to be updated with information on availability? Message 6 of
|Date Added:||25 January 2011|
|File Size:||50.58 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The driver only uses the Fetch and Read commands so I can’t hp34401x that the Measure command is correct. Install the driver and follow the steps in my original answer.
You’ll get there quickly. Message 6 of Remote operation of the instrument usually: For the measurements, go back a couple of steps hp344011a select “Measurement”, there are a couple of selections to choose from.
Message 5 of You have to put together the building blocks to get your measurement. Let’s see if I can get you on glib way with the following: The A for the next generation is the A providing the same uncompromising performance for system and benchtop testing along with new capabilities you need.
Here is a link for the driver: With the above named VI you first send kind of an initialization to the instrument: I have the HPA Init.
The recommended calibration interval for these devices is 12 months.
Keysight Technologies / Agilent Technologies hp34401a Meter
You’ve got some code there only for the assuming the visa referecne is correct and you’re asking for the voltage not current. Thank you Very much. Right after that you read the answer from your instrument gppib the GPIB Hope that is of any help for you. Hello Joe and Anja, Thank for your help.
You can now ask the instrument for the currently measured data i. Message 3 of Thank you very much for your attention.
So try the following: I don’t understand the attachment. I followed your conversation with Joe Message 8 of Just call us, we will help you to find the right solution! I also sometimes ignore existing drivers and create my own.
Of course you can add some kind of control of the communication if you want. Message 4 of Go to the block diagram.
EOL/obsolete: Keysight A (HPA) multimeter
If you continue browsing the site you consent to our cookie use. Message 10 of Start with HPA Init. Thank for your help.